To avoid e lectric shock, donot o perate your meter until the
battery door i sinplace a nd fastened
To avoid e lectric shock, disconnect
the test l eads from any
source of voltage before removing the fuse door.
1. Disconnect the test leads from the meter and any item under test.
Open the fuse door by loosening the screw on the door using a Phillips
Remove the old fuse from its holder by gently pulling it out.
Install the new fuse into the holder.
5. Always use a fuse of the proper size and value (0.5A/250V fast blow
for the 400mA range, 10A/250V fast blow for the 10A range).
Put the fuse door back in place. Insert the screw and tighten it securely.
To avoid electric shock, do not operate your meter unti. the]
fuse door is in place and fastened securely.
The UL mark does not indicate that this product has been evaluated for the
accuracy of its readings.